The Laryngeal Mask Airway SupremeTM – a single use laryngeal mask airway with an oesophageal vent. A randomised, cross-over study with the Laryngeal Mask Airway ProSealTM in paralysed, anaesthetised patients
Eschertzhuber, S., Brimacombe, J., Hohlrieder, M., and Keller, C. (2009) The Laryngeal Mask Airway SupremeTM – a single use laryngeal mask airway with an oesophageal vent. A randomised, cross-over study with the Laryngeal Mask Airway ProSealTM in paralysed, anaesthetised patients. Anaesthesia, 64 (1). pp. 79-83.
|PDF (Published Version) - Repository staff only - Requires a PDF viewer such as GSview, Xpdf or Adobe Acrobat Reader|
View at Publisher Website: http://dx.doi.org/10.1111/j.1365-2044.20...
The LMA SupremeTM is a new extraglottic airway device which brings together features of the LMA ProSealTM, FastrachTM and UniqueTM. We test the hypothesis that ease of insertion, oropharyngeal leak pressure, fibreoptic position and ease of gastric tube placement differ between the LMA ProSealTM and the LMA SupremeTM in paralysed anesthetised patients. Ninety-three females aged 19–71 years were studied. Both devices were inserted into each patient in random order. Two attempts were allowed. Digital insertion was used for the first attempt and guided insertion for the second attempt. Oropharyngeal leak pressure and fibreoptic position were determined during cuff inflation from 0 to 40 ml in 10 ml increments. Gastric tube insertion was attempted if there was no gas leak from the drain tube. First attempt and overall insertion success were similar (LMA ProSealTM, 92% and 100%; LMA SupremeTM 95% and 100%). Guided insertion was always successful following failed digital insertion. Oropharyngeal leak pressure was 4–8 ml higher for the LMA ProSealTM over the inflation range (p < 0.001). Intracuff pressure was 16–35 cm higher for the LMA ProSealTM when the cuff volume was 20–40 ml (p < 0.001). There was an increase in oropharyngeal leak pressure with increasing cuff volume from 10 to 30 ml for both devices, but no change from 0 to 10 ml and 30–40 ml. There were no differences in the fibreoptic position of the airway or drain tube. The first attempt and overall insertion success for the gastric tube was similar (LMA ProSealTM 91% and 100%; LMA SupremeTM 92% and 100%). We conclude that ease of insertion, gastric tube placement and fibreoptic position are similar for the LMA ProSealTM and LMA SupremeTM in paralysed, anaesthetised females, but oropharyngeal leak pressure and intracuff pressure are higher for the LMA ProSealTM.
|Item Type:||Article (Refereed Research - C1)|
|FoR Codes:||11 MEDICAL AND HEALTH SCIENCES > 1103 Clinical Sciences > 110301 Anaesthesiology @ 100%|
|SEO Codes:||92 HEALTH > 9201 Clinical Health (Organs, Diseases and Abnormal Conditions) > 920118 Surgical Methods and Procedures @ 100%|
|Deposited On:||25 Sep 2009 11:08|
|Last Modified:||30 Apr 2013 00:49|
Last 12 Months: 0
|Citation Counts with External Providers:||Web of Science: 44|
Repository Staff Only: item control page